Main Article Content
The innovation of JVP meter aimed to develop of nursing innovation. The purposes of this study were to develop an instrument to examine jugular venous pressure and to investigate the satisfaction of registered nurses toward the use of the JVP meter. The development of the JVP meter was divided into two main steps. The first step was to develop the JVP meter in accordance with PDCA cycle: 1) plan and study concept and related studies; 2) do; 3) check; and 4) act by evaluating and reporting results and suggestions as a guideline for further study. The second step was to investigate the satisfaction of registered nurses toward the use of the JVP meter. The samples were selected using purposive sampling which consisted of 2 registered nurses who working at the Heart Failure Clinic, Sakon Nakhon Hospital over 10 years, and 30 patients with heart failure. The data were analyzed by using inter-rater reliability: IRR, content analysis, mean, standard deviation (SD.), frequency, and percentage.
The results of the study were as follows: 1) the inter-rater reliability: IRR was “Very Good” (K ≥0.96; p= 0.01) and 2) the overall satisfaction of using the JVP meter was “High” (Mean = 4.29, SD = 0.38). In particularly, the strength and durability aspect and reasonable price and production cost aspect was at a “Highest” level of satisfaction.
Article published Is the copyright of the Journal of Health and Nursing Research (Boromarajonani College of Nursing, Bangkok) Cannot be republished in other journals
editor. Comprehensive Heart Failure Management Program. Chiangmai: Chiangmai University; 2014. P 77-92. (in Thai).
2. Phrommintiku A. Symptoms and signs of Heart failure. In Kanjanavanit R & Phrommintiku A, editor. Comprehensive Heart Failure Management Program. Chiangmai: Chiangmai University; 2014. P 5-20. (in Thai).
3. Promnoi P. Central Venous Pressure Monitoring: Clinical Practice Guide for Nurses. Songklanagarind Journal of Nursing 2012;32(1):45-52. (in Thai).
4. Micheal J, Parihn N, Fergusson D. The jugular venous pressure revisited. Level and Clinic Journal of Medicine 2013;80(10):638–44.
5. Bhattarai M. Prospects of jugular venous pulse assessment. Journal of Nepal Medical Association
6. Naveen G, Nitish G. Jugular Venous Pulse: An Appraisal Clinical medicine. Journal Indian Academy of clinical medicine. 2000;1(3):260-9.
7. Jevon P. cardiovascular examination Measuring jugular venous pressure. Clinical practical
8. Socransky SJ, Wiss R, Robins R, Anwati A, Roy M, Yeung C. Defining normal jugular venous pressure with ultrasonography. Canadian journal of emergency medicine 2010;12(4):320-24.
9. University of Washington Department of medicine. Techniques: Jugular venous pressure
Measurement [Internet]. 2013 [cited 2015 Sep 15]. Available from https://depts.washington.
10. Schmitz A, Moehling T, Harris K, Horn D. Bedside device to measure the jugular venous pressure [Internet]. 2012 [cited 2015 Sep 15]. Available from http://bmedesign.engr.wisc.edu/projects/file/?fid=2452.
11. Phasunon P. Evaulation on Inter-Rater Reliability Using Kappa Statistics. The Journal of Faculty of Applied Arts 2015;8(1):2-20. (in Thai).
12. Thomjorhor N. Ruler Measurement [Internet]. 2016 [cite 2016 Aug 9]. Available from: http://
tonanasia.com/wordpress/techniques/technical -measurement/. (in Thai).